Helpful discussions have been held with N. Larson and D. Murphy of Sea-Bird Electronics. We specially thank N. Larson for his particularly interesting information regarding the environmental conditions and instrument configurations used for Sea-Bird testing, as well as helpful advice. The results we have found and generally all the works presented here would not have been possible without the crews and SHOM hydrographer teams who acquired or helped to acquire the numerous data used in this study.
Bray, N. A., 1987: Salinity calculation techniques for separately digitized fast response and platinum resistance CTD temperature sensors. Deep-Sea Res., 34 , 627–632.
Gregg, M. C., , and Hess W. C. , 1985: Dynamic response calibrations of Sea-Bird temperature and conductivity probes. J. Atmos. Oceanic Technol., 2 , 304–313.
Gregg, M. C., , Schedvin J. C. , , Hess W. C. , , and Meagher T. B. , 1982: Dynamic response calibrations of the Neil Brown conductivity cell. J. Phys. Oceanogr., 12 , 720–742.
Horne, E. P., , and Toole J. M. , 1980: Sensor response mismatches and lag correction techniques for temperature-salinity profilers. J. Phys. Oceanogr., 10 , 1122–1130.
Johnson, G. C., , Toole J. M. , , and Larson N. G. , 2007: Sensor corrections for Sea-Bird SBE-41CP and SBE-41 CTDs. J. Atmos. Oceanic Technol., 24 , 1117–1130.
Lueck, R. G., , and Picklo J. J. , 1990: Thermal inertia of conductivity cells: Observations with a Sea-Bird cell. J. Atmos. Oceanic Technol., 7 , 756–768.
Morison, J., , Andersen R. , , Larson N. , , D’Asaro E. , , and Boyd T. , 1994: The correction for thermal-lag effects in Sea-Bird CTD data. J. Atmos. Oceanic Technol., 11 , 1151–1164.
Ochoa, J., 1989: A practical determination of CTD platinum resistance thermometer response time, and its use to correct salinity bias and spikes. Deep-Sea Res., 36 , 139–148.
Schmitt, W., , Millard R. C. , , Toole J. M. , , and Wellwood W. D. , 2005: A double-diffusive interface tank for dynamic-response studies. J. Mar. Res., 63 , 263–289.